Institute for Public Health, Ministry of Health Malaysia, Kuala Lumpur, Malaysia
*Corresponding author: Noor Ani Ahmad. Institute for Public Health, Ministry of Health Malaysia, Jln Bangsar, Kuala Lumpur, Malaysia. E-mail: email@example.com
Published: June 25, 2015. DOI: 10.21103/Article5(2)_MME1
Background: Point-of-care testing, POCT, was widely used to assess hemoglobin status before proceeding with the confirmation test. Our study aimed to assess the validity of Mission® Plus Hb in detecting hemoglobin levels in a general population compared with a standard laboratory hematology analyser as the gold standard.
Methods: Two types of samples, capillary and venous blood, were collected from all respondents by trained nurses. Both blood samples were tested using Mission® Plus while the remaining venous blood in EDTA test tubes was sent to the reference laboratory.
Results: A total of 622 respondents participated in this study, 75% of them females. The mean Hb tested from capillary blood using Mission® Plus Hb was 11.80±2.02g/dl. For venous blood, the mean Hb concentrations using Mission® Plus Hb and Sysmex XE-2100 hematology analyser were 12.16±1.84g/dl and 13.07±1.87g/dl, respectively. Sensitivity and specificity in detecting anemia from venous samples were 98.8% and 73.4% respectively. Positive Predictive Value (PPV) was 58.5%, while Negative Predictive Value (NPV) was 99.4%.
Conclusion: The findings of moderate PPV should alert the programme managers to the importance of a confirmatory test following screening using Mission® Plus Hb.
- Worldwide prevalence of anaemia 1993–2005: WHO global database on anaemia. Edit by De Benoist B, McLean E, Egli I, Cogswell M. World Health Organization, Geneva, 2008.
- Sagin DD, Ismail G, Mohamad M, Pang EK, Sya OT. Anemia in remote interior communities in Sarawak, Malaysia. Southeast Asian J Trop Med Public Health 2002; 33(2): 373-7.
- Hassan R, Abdullah WZ, Nik Hussain NH. Anemia and iron status of Malay women attending an antenatal clinic In Kubang Kerian,Kelantan, Malaysia. Southeast Asian J Trop Med Public Health 2005; 36(5):1304-7.
- Thirukkanesh S, Zahara AM. Compliance to Vitamin and Mineral Supplementation among Pregnant Women in Urban and Rural Areas in Malaysia. Pakistan J Nutrition 2010; 9(8):744-750.
- Srivastava T, Negandhi H, Neogi SB, et al. Methods for hemoglobin estimation: a review of “What Works”. J Hematol Transfus 2014; 2(3):1028.
- Sanchis-Gomar F1, Cortell-Ballester J, Pareja-Galeano H, Banfi G, Lippi G. Hemoglobin point-of-care testing: the HemoCue system. J Lab Autom 2013; 18(3):198-205.
- Neufeld L, García-Guerra A, Sánchez-Francia D, Newton-Sánchez O, Ramírez-Villalobos MD, Rivera-Dommarco J. Hemoglobin measured by Hemocue and a reference method in venous and capillary blood: a validation study. Salud Publica Mex 2002; 44(3):219-27.
- Paiva Ade A, Rondó PH, Silva SS, Latorre Mdo R. Comparison between the Hemocue and an automated counter for measuring hemoglobin. Rev Saude Publica 2004; 38(4):585-7.
- Shahshahani HJ, Meraat N, Masouri F. Evaluation of the validity of a rapid method for measuring high and low haemoglobin levels in whole blood donors. Blood Transfus 2013; 11(3):385-90.
- Sample size calculation for sensitivity and specificity studies. http://www.docstoc.com/docs/168682724/Sample-size-for-sensitivity-_-spec... (accessed on January 5, 2015).
- Walters J, Garrity P. Performance Evaluation of the Sysmex XE-2100 Hematology Analyzer. Lab Hematol 2000; 6:83-92.
- Tsuruda K, Tsuji T, Usui T, et al. Evaluation and Clinical Usefulness of the Automated Hematology Analyser, Sysmex XE-2100. Sysmex J Int 1999; 9(2):129-138.
- National Point of Care Testing: Policy and Guidelines. Departmental Policy of Pathology Services, Ministry of Health Malaysia, 2012.
- Gehring H, Hornberger C, Dibbelt L, Rothsigkeit A, Gerlach K, Schumacher J, et al. Accuracy of point-of-care-testing (POCT) for determining haemoglobin concentrations. Acta Anaesthesiol Scand 2002; 46(8):980–6.
- Agarwal R, Heinz T. Bedside hemoglobinometry in hemodialysis patients: lessons from point-of-care testing. ASAIO J 2001; 47(3):240-3.
- Nguyen HT. High humidity affects HemoCue cuvette function and HemoCue haemoglobin estimation in tropical Australia. J Paediatr Child Health 2002; 38(4):427-8.
- Mendrone A Jr1, Sabino EC, Sampaio L, Neto CA, Schreiber GB, Chamone Dde A, et al. Anemia screening in potential female blood donors: comparison of two different quantitative methods. Transfusion 2009; 49(4):662-8.
- Akhtar K, Sherwani RK, Rahman K, Hasan J, Shahid M.Hemocue photometer: a better alternative of haemoglobin estimation in blood donors? Indian J Hematol Blood Transfus 2008; 24(4):170-2.
- Bahadur S, Jain S, Jain M. Estimation of hemoglobin in blood donors: a comparative study using hemocue and cell counter. Transfus Apher Sci 2010; 43(2):155-7.
- Cable RG, Steele WR, Melmed RS, Johnson B, Mast AE, Carey PM, et al. The difference between fingerstick and venous haemoglobin and hematocrit varies by sex and iron stores. Transfusion 2012; 52(5):1031-40.
The fully formatted PDF version is available.
Int J Biomed. 2015; 5(2):91-94. © 2015 International Medical Research and Development Corporation. All rights reserved.